|Place of Origin:||CHINA|
|Minimum Order Quantity:||Negotiation|
|Payment Terms:||T/T, L/C|
|Supply Ability:||100000 sets per year|
|Operating Mode:||Ablation, Coagulation, Hemostasis||Output Energy:||Adjustable|
|Application:||Tonsillectomy, Adenoidectomy, Soft Palate||Advantage:||Minimally Invasive, Low Temperature, Quick Revocery|
rf ablation system,
rf ablation devices
Minimally Invasive Snoring Treatment System ENT Plasma Surgery System Coblator
MECHAN has been working to develop advance Plasma technology into much more medical applications. OSAHS has been recognized as a personel and social problem, our ENT plasma surgery system PLA-700 are made to solve these problems.
|1 Plasma Generator||2 Power Cord||3 Footswitch Plate||4 ABLATION Control|
|5 PLACOAG Control||6 Plasma Probe||7 Receptacle of Footswitch||8 Receptacle of Handle|
|Input Volts||198-240V 50Hz
Export device could convert automatically per customer requirements
|Output Power||Ablation or resection: ≤351W
Coagulation or hemostasis: ≤111W
|Working Frequency||100 KHz ± 10%|
|Working Mode||Ablation and resection: level 1-10
Coagulation: level 1-10
A 50 μm plasma area is formed around the electrode through a conductive medium (mainly H2O, NaCl and air). Unlike traditional electrosurgical methods, which separate and solidify tissues by heating the tissue, Metron plasma knives use the kinetic energy generated by the rapid movement of plasma in an electric field at 100 kHz to decompose the tissue, thereby achieving the purpose of cutting and clotting.
System Set Up
1). One generator
2). One reusable and non-sterile power supply cable.
3). One reusable and non-sterile footswitch.
4). One reusable and non-sterile treatment handle which is for foot control.
5). Treatment probes.
|Manufacturer Logo||MECHEN Brand.|
|Pads||Insulation and support on the surface.|
|Receptacle of Footswitch||Connects with Footswitch which determines ablation or coagulation mode.|
|Receptacle of Treatment Handle||Connects with the treatment accessories.|
|Power Switch||System start power button. “O” means power off; “I” means power on.|
|ABLATION mode Selector||Select ABLATION mode 1 or 2 Output waveform.|
|Indicator of Treatment Probe Connection||LED turns on green when the probe cable is connected and turns on red when it is not connected.|
|Indicator of Footswitch Connection||LED turns on green when the footswitch cable is connected and turns on Red when it is not connected.|
|ABLATION Power Setting Display||Shows the preset ABLATION output value in two digits.
“01” with Min. power output, and “10” with the highest output.
|Ablation Output Power Up/Down Button||The number displayed indicates the ablation nominal power, in watts, which will be delivered to the patient when the mode is activated. Only available for changing in ready mode.|
ENT System Features
1. Push-button operation, easy to use;
2. Real-time monitoring of the tissue impedance value during the surgery to tell the degree of surgery;
3. peeling cutting probe(electrode), cutting is more convenient and rapid;
4. more surgical electrodes to choose to meet more parts of tissue ablation, cutting, coagulation and hemostasis;
5. ablation cutting mode 1, ablation cutting mode 2, to meet the ablation cutting needs of different organizations.
Scope of Adaption
The plasma surgery system is used in conjunction with the disposable plasma probes, suitable for ablation, cutting, coagulation and hemostasis of human tissue. It needs to be used in a saline environment.
This device should be stored in a well-ventilated room without corrosive gas. The height above ground and the distance from wall should be above 30cm. Stack limit is 6 cases. Before being stored for a long time, it should be cleaned thoroughly for proper packing and storage. And it should also be taken out for power up every year, in order to avoid it from moisture, mildew or even damage. The suitable weather and environment condition is as below:
1. Temperature: -40℃～+55℃;
2. Relative humidity: ≤90％;
3. Atmospheric pressure: 50kPa～106kPa
Contact Person: Ivy