|Place of Origin:||CHINA|
|Minimum Order Quantity:||Negotiation|
|Payment Terms:||T/T, L/C|
|Supply Ability:||100000 sets per year|
|Ablation:||Adjustable From 1 To 10||Coagulation:||Adjustable From 1 To 10|
|Application:||Tonsillectomy, Adenoidectomy, CAUP, UPPP||Advantage:||Minimally Invasive, Low Temperature|
|Warranty:||24 Months After Sterilization||Shaft Length:||120mm|
|Usage:||Ablation, Cutting, Resection, Hemostasis||Feature:||Disposable|
ent surgical instruments,
portable plasma wand
Tonsillectomy Plasma Wand ENT Probe For Snoring Treatment System
PLA401 is design specially for Tonsillectomy, Adenoidectomy, CAUP, UPPP treatment. It is suitable for the length of the blade required for tonsil and adenoid surgery. During adenoidectomy, it is easier to access the hyperplasia tissue in the posterior nostril, and the blade is easier to bend. It has the function of cutting, ablation, hemostasis and coagulation. It is connect with PLA-700 ENT plasma surgery system to perform the operation.
Product Tip Picture
System Set Up
|Manufacturer Logo||MECHEN Brand.|
|Receptacle of Footswitch||Connects with Footswitch which determines ablation or coagulation mode.|
|Receptacle of Treatment Handle||Connects with the treatment accessories.|
|Power Switch||System start power button. “O” means power off; “I” means power on.|
|ABLATION mode Selector||Select ABLATION mode 1 or 2 Output waveform.|
|Indicator of Treatment Probe Connection||LED turns on green when the probe cable is connected and turns on red when it is not connected.|
|Indicator of Footswitch Connection||LED turns on green when the footswitch cable is connected and turns on Red when it is not connected.|
|ABLATION Power Setting Display||
Shows the preset ABLATION output value in two digits.
“01” with Min. power output, and “10” with the highest output.
|Ablation Output Power Up/Down Button||The number displayed indicates the ablation nominal power, in watts, which will be delivered to the patient when the mode is activated. Only available for changing in ready mode.|
Relying on the mechanical kinetic energy of the plasma in the electric field, it hits the molecular bonds (40-70 ° C) of the target tissue to lyse and ablate. The thermal damage depth is less than 0.5mm, and the trauma to normal adjacent tissues is small.
Tonsillectomy, Adenoidectomy, CAUP, UPPP
Electrodes shall be stored in environment of humidity less than 80%. no corrosive gas, well vented room. Taking complete protection to electrodes.
Contact Person: Ivy