|Place of Origin:||CHINA|
|Minimum Order Quantity:||10PCS|
|Delivery Time:||1-3 weeks|
|Supply Ability:||100000 per month|
ent surgical instruments,
portable plasma wand
Plasma ENT Probe Minimally Invasive Surgery for Turbinate Channeling Wand
Setting Up the unit:
Verify that the Power On/Off Switch is in the Off position and that no accessories are connected to the unit.
Connect a hospital grade power cable to the AC power cable receptacle on the back of the unit, then to a properly grounded wall outlet.
Turn the unit on by switching the power switch to the On position.
Connect a two-pedal footswitch to the appropriate receptacle on the rear of the unit. Use only MECHAN’s footswitches. Although other types of footswitches may fit, they may not be compatible.
Connect a treatment handle and electrode at this time
Fire Explosion Hazard:
The following substances will contribute to increased fire and explosion hazards in the operating room:
• Flammable substances (such as alcohol based skin prepping agents and tinctures)
• Naturally occurring flammable gases which may accumulate in body cavities such as the bowel
• Oxygen enriched atmospheres
• Oxidizing agents (such as nitrous oxide [N2O] atmospheres)
The sparking and heating associated with electric surgery can provide an ignition source. Observe fire precautions at all times. When using electric surgery in the same room with any of these substances or gases, prevent their accumulation or pooling under surgical drapes, or within the area where electric surgery is performed.
Contact Person: plasmasurgery