|Place of Origin:||China|
|Minimum Order Quantity:||1 SET|
|Packaging Details:||Wooden Box|
|Payment Terms:||T/T, L/C, Western Union|
|Supply Ability:||100 units per day|
|Applying Field:||Urology Department||Color:||Orange|
|Net Weight:||7KG||Compatible Probe:||PLA201, PLA202, PLA204, PLA205, PLA207|
|Warranty:||2 Years||Packing:||Paper Box|
utm testing machine,
digital utm machine
Urology Plasma Surgery System for BPH Treatment
This PLA-600 Model Urology Plasma Surgery System are designed to appliy into the field of urology. On the front panel, there are two working modes, Ablation mode and Coagulation mode. Which can be adjust from 1 to 10 level by push the plus + or reduce - button. It is widely applying to BPH and Bladder tumor treatment.
Export device could convert automatically per customer requirements
Ablation or resection: ≤351W
Coagulation or hemostasis: ≤111W
|Working Frequency||100 KHz ± 10%|
Ablation and resection: level 1-10
Coagulation: level 1-10
System Set Up
|Manufacturer Logo||MECHEN Brand.|
|Pads||Insulation and support on the surface.|
|Receptacle of Footswitch||Connects with Footswitch which determines ablation or coagulation mode.|
|Receptacle of Treatment Handle||Connects with the treatment accessories.|
|Power Switch||System start power button. “O” means power off; “I” means power on.|
|ABLATION mode Selector||Select ABLATION mode 1 or 2 Output waveform.|
|Indicator of Treatment Probe Connection||LED turns on green when the probe cable is connected and turns on red when it is not connected.|
|Indicator of Footswitch Connection||LED turns on green when the footswitch cable is connected and turns on Red when it is not connected.|
|ABLATION Power Setting Display||
Shows the preset ABLATION output value in two digits.
“01” with Min. power output, and “10” with the highest output.
|Ablation Output Power Up/Down Button||The number displayed indicates the ablation nominal power, in watts, which will be delivered to the patient when the mode is activated. Only available for changing in ready mode.|
1). One tabletop generator device
2). One reusable and non-sterile power supply cable.
3). One reusable and non-sterile footswitch.
4). One reusable and non-sterile treatment handle which is for foot control.
5). Treatment probes.
This device should be stored in a well-ventilated room without corrosive gas. The height above ground and the distance from wall should be above 30cm. Stack limit is 6 cases. Before being stored for a long time, it should be cleaned thoroughly for proper packing and storage. And it should also be taken out for power up every year, in order to avoid it from moisture, mildew or even damage. The suitable weather and environment condition is as below:
1. Temperature: -40℃～+55℃;
2. Relative humidity: ≤90％;
3. Atmospheric pressure: 50kPa～106kPa
Contact Person: Ivy